[product name] reinforced tracheal intubation。
【 specification 】 2.0 # # #, # 2.5, 2.5, 3.5, 4.0, # # #, # 4.5, 4.5, 5.5, 6.0, # # #, # 6.5, 6.5, 7.5, 8.0, # # #, # 8.5, 8.5, 9.5, 10.0, # #, # 10.5 and 10.5
[name of registrant] henan jianqi medical equipment co. LTD.
The dingluan industrial zone in changyuan county.
[contact number] 0373-8690789.
[after-sales service unit] henan jianqi medical equipment sales co。, LTD。
[name of production enterprise] henan jianqi medical equipment co., LTD.
The dingluan industrial zone in changyuan county。
[production address] dingluan industrial zone, changyuan county.
[contact information] 0373-8690789。
[production license] the food and drug regulatory apparatus is permitted to be made in 20080036.
[registration certificate no.] yu ordnance note 20172660189.
[technical requirements no。]
[structure and composition] the product is composed of stainless steel tube body, sleeve, joint, inflatable device and guide core.
[product performance] 1. Strengthen the pipe machine end joint and the patient end surface should be neat and smooth, without burr, without injection flow; Strengthen the tube body extrusion even without glue flow, the stainless steel tube should be symmetrical in the middle of the wall. 2. The basic size of the reinforced pipe shall be in accordance with the provisions of YY 0337.1-2002. 3. The performance of the reinforced pipe shall be in accordance with the provisions of YY 0337.1-2002. 4. Strengthen the negative pressure of the steel tube with steel tube in 40Kpa, and continue with 15s. 5. The guide core shall be clean, smooth and free of impurities, and can be molded. 6. The PH value should be 6.5 ~ 7.5. 7. Products should be sterile. 8. With the enhanced tube of ethylene oxide sterilization, the residual amount of ethylene oxide should not exceed 10 mu g/g.
[scope] for the establishment of artificial airway in clinical anesthesia or first aid.
湖北11选5开奖【 forbidden disease 】 allergic to product materials, laryngeal edema and acute inflammation of the airway and throat abscess, oppressed trachea thoracic aortic aneurysms, severe bleeding constitution patients disabled, unless first aid, emergency endotracheal intubation.
1。 This product is a disposable product, which is damaged in packaging and is strictly prohibited to be used and destroyed after use。
2. See packaging seal or label on the production batch number, and the sterilization date is in the outer packing box.
湖北11选5开奖3. This product is sterilized by ethylene oxide.
[use method] 1. Select the applicable patient's specifications and model, and then open the package after the outer packing is in good condition.
2。 Lie on the back of the patient, check the oral cavity, (remove foreign body and active teeth, and fall without tongue)。
3。 Use the laryngoscope to provoke the patient's epiglottis to expose the glottis, and insert the organ intubation on the side of the laryngoscope。
4. Press the patient's chest with air flow to indicate the success of intubation.
5. After the product is finished, it should be treated with medical waste to avoid pollution.
[storage method] 1. This product is used in the carriage and hold of the cabins, and is kept clean, without pressure, sunshine and rain and snow.
2. Handle gently in handling and avoid violent collision;
3. The product shall be stored away from the fire source, and the relative humidity shall not exceed 80%, without corrosive gas and well-ventilated and clean environment.
[production date] see packaging seal or label。
湖北11选5开奖Expiry date: see packaging seal or label.
[label, package logo]
[date of instruction] May 3, 2017.